This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||18 June 2015|
|PDF File Size:||20.25 Mb|
|ePub File Size:||7.56 Mb|
|Price:||Free* [*Free Regsitration Required]|
Table of Contents Add to my manuals Sc1000. Setting Up System Security Operation 4. Don’t show me this message again. To locate the representative nearest you, send e-mail to intl hach. Troubleshooting Section 7 Troubleshooting Troubleshooting Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions. Refer to Figure 29 and perform the following steps to replace the fuses: Index Numerics 90 Degree Detector Page 23 Allow at least mm In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase hah, subject to the pro-rated sc100 above, excluding original shipping and handling charges.
Got it, continue to print.
To print the manual completely, please, download it. In the United States: The event log stores a variety of events that occur on the devices such xc100 configuration changes, alarms, and warning conditions. Controller Dimension Illustrations Installation 3.
Index Index Numerics 90 Degree Detector Seat insulation against connector with no bare wire exposed. Wire gauge less than 18 AWG is not recommended. Page 20 Installation 3.
Hach-Lange SC 100 User Manual
Printed in the U. Relay loads must be resistive. Page 72 Visit http: Wet Verification System Startup 5.
Please contact the Hach Service Center serving your location. Section 7 Troubleshooting 7.
Page 71 Index Sample Line Installation Remove the quick connect fitting and wires and install the threaded plug on the opening to hachh the environmental rating. The incident light Method of Detection beam is composed of a tungsten filament light source with a color temperature between and K. The display will flash on startup, when the hold outputs function has been activated, and when the filter function signal average is changed to a different value.
System Startup Two types of verifications are offered. In addition, the passcode also regulates function selection for sv100 options.
Sensor Setup Menu Operation 4. Installation Electrical Installation High-voltage wiring for the controller is conducted behind the high voltage DANGER The instrument must be installed barrier in the controller enclosure.
By pressing ‘print’ button you will print only current page. Section 10 Repair Service Authorization must be obtained from Hach Company before sending any items for repair.
Index Sample Line Installation Close and secure the cover. Restore sample flow to the instrument.
HACH SC E INSTRUMENT MANUAL Pdf Download.
The calibration history log shows the sc10 value, the time and date of the calibration, and the initials of the operator performing verification.
Enter text from picture: Don’t have an account? Inadequate bubble removal from Increase the signal averaging time to a longer interval.
Section 5 System Startup 5. Page of 72 Go.
Hach-Lange SC User Manual | 58 pages
Page 35 Operation 4. Changing The Sensor Name Operation 4. Replacing The Lamp Assembly Section 2 General Information 2. Pull the lamp leads and connector through the hole in the lamp housing.
To hold the output until released: